Thanks for the replys.
I think I will go ahead and move to Fluke. Mabey get the UEI DT200 warrantied and eBay it. I wished Fluke 902 had the dual display like the UEI DL389 I have now, and I especially like the magnet on the back of the UEI, but it has caused me some unnecessary problems:
1. The test leads that come with the meter are crap, (even though they do snap neatly into the back of the meter), on my third set in under two years. First bad set led me to wrongly replace a good control transformer.
2. Reading 300 something volts on a large condenser actually running about 490; would have looked like a real dumb ass if I had called power company without checking with another meter.
3. Shows 35 amps while reading two 20amp heat strips together. Another meter read 40 like it should have. Both problems corrected by replacing battery when low battery indicator was not on.
So I guess I will sacrifice my magnet and dual display, and get a Fluke before the UEI waste any more of my time or kills me.