I am trying to monitor metering using an NAE and DX9100, have mapped in the DI as ADI onto the NAE, counts up fine, I wanted to produce trend studies for these meters so trended the ADI point but register only counts to 32,000 (or there abouts) then resets to 0. Not such a good graph profile!!
I now am trying to use a totaliser extension, with a control system reseting the totaliser after an hour and taking a tend of the output, BUT the totaliser appears to continue to increase when the meter is not pulsing. Any ideas anyone, its sending me mad. Thanks