I have a many HOBO data loggers, model type U12-013, and I noticed that the internal 10k thermistor is suspect in comparison to the two plug in thermistors.
My question is, can these data loggers be calibrated? By Hoboware, if so, how is this done with software? Or sending them to Onset, if so, how long does this take and is it cost effective (i.e. should I just buy a new logger)?
HOBO logger Model: