Type: Posts; User: jimp
Harbor freight has a nice pump for $89.
Taking credit cards is unprofessional!
Yea man, that Chinese chrome looks good!
I am always on call and it pays nothing. But If I go out it pays $187.50 per hour minimum 4hrs plus travel exp.
I rent a Portable 10HP MRI chiller on a trailer that has all the necessary fittings and hoses for temporary hookup. It has never been a question of "how much" when they are looking at the loss of...
I have the 2011 Sprinter and I bought it new, no problems at 49K. Changed over from a E350 KUV and my mileage nearly doubled. Before the KUV I owned Ford vans since the 70's.
After owning the...
Equalize the static on both sides of the system.
So U checked the ohms?
Myself I only work Cascade - Auto Cascade and some MR Helium stuff. 95% is inside conditioned space with no ladder work. So I guess that makes me a candy A$$ like the guy with the Orange truck said.
I had a leak last week on the defrost solenoid for the misner on a PFC/PFC 1100 ST. It was a pin hole on the stem under the coil, the machine lost about 3 to 4 psi a month static.
The leak before...
Working outside is bad on your health, inside career is the way to go.
Anyone here care to share your experiences on where you have found leaks on the 600 through 1100 series systems.
I have a 35L Ultra EL-85 thats dropping output #3 in 4 day recipe on the Allen Bradley. Output #3 controls the solenoid valve for condenser cooling. After #3 drops output of course you lose...
Don't use gauges, use a pump manifold with 1/4" copper for hoses.
A better picture of the SG, I don't see any signs that it was overheated.
Sight glass is a factory install, no signs of overheating. The system is clean, also I did a blow through acid test which was negative.
Over the years I have run across sight glasses that lose the eye. Is their any particular reason why this happen?
95% of my jobs are MR. Its a great field if you have a complete understanding of what goes on beyond the chiller.
Changed a few of those back in the day, they had water cooled condenser's wrapped around them.
We called them footballs.
Can I get a free post?
The shelf compressor only cools the shelf, it has two caps. One cap is for cooling, one is for when the recipe calls for no cooling. The condenser compressor cools the condenser of course, unless...
One of the machines is a Virtis Ultra LE 25, has daul compressors with receivers and caps.
Funny, I have several factory build systems that run from -60c to -150c with caps and receivers.
Hands, drain the system and fill with fresh water, add a product called Citranox for cleaning. I try to run the cleaner for 24hrs at a elevated temperature for the best results. After its clean pull...